Implants based on the polyetheretherketon (PEEK) polymer have been developed in the last decade as an alternative to conventional metallic devices. PEEK devices may provide several advantages over the use of conventional orthopedic materials, including the lack of metal allergies, radiolucency, low artifacts on magnetic resonance imaging scans and the possibility of tailoring mechanical properties. The purpose of this study was to evaluate the clinical results at 12-month follow-up using a new plate made of carbon-fiber-reinforced polyetheretherketon for the treatment of distal radius fractures.
We included 40 consecutive fractures of AO types B and C that remained displaced after an initial attempt at reduction. The fractures were classified according to the AO classification: 21 fractures were type C1, 9 were type C2, 2 were type C3, 2 were type B1 and 6 were type B2.
At a 12-month follow-up no cases of hardware breakage or loss of the surgically achieved fracture reduction were documented. All fractures healed, and radiographic union was observed at an average of 6 weeks. The final Disabilities of Arm, Shoulder and Hand score was 6.0 points. The average grip strength, expressed as a percentage of the contralateral limb, was 92 %. Hardware removal was performed only in one case, for the occurrence of extensor tenosynovitis.
At early follow-up this device showed good clinical results and allowed maintenance of reduction in complex, AO fractures.
Akay M, Aslan N (1995) An estimation of fatigue life for a carbon fibre/poly ether ether ketone hip joint prosthesis. Proc Inst Mech Eng [H] 209:93–103 CrossRef
Skinner HB (1998) Composite technology for total hip arthroplasty. Clin Orthop Relat Res 235:224–236
Steinberg EL, Rath E, Shlaifer A, Chechik O, Maman E, Salai M (2013) Carbon fiber reinforced PEEK Optima—a composite material biomechanical properties and wear/debris characteristics of CF-PEEK composites for orthopedic trauma implants. J Mech Behav Biomed Mater 17:221–228. doi: 10.1016/j.jmbbm.2012.09.013 PubMedCrossRef
Arora R, Lutz M, Deml C, Krappinger D, Haug L, Gabl M (2011) A prospective randomized trial comparing nonoperative treatment with volar locking plate fixation for displaced and unstable distal radial fractures in patients sixty-five years of age and older. J Bone Joint Surg Am 93:2146–2153. doi: 10.2106/JBJS.J.01597 PubMed
Slätis P, Karaharju E, Holmström T, Ahonen J, Paavolainen P (1978) Structural changes in intact tubular bone after application of rigid plates with and without compression. J Bone Joint Surg Am 60:516–522 PubMed
Huiskes R, Weinans H, van Rietbergen B (1992) The relationship between stress shielding and bone resorption around total hip stems and the effects of flexible materials. Clin Orthop 274:124–134 PubMed
- A new volar plate made of carbon-fiber-reinforced polyetheretherketon for distal radius fracture: analysis of 40 cases
- Springer International Publishing
Neu im Fachgebiet Orthopädie und Unfallchirurgie
Mail Icon II