A wide application of ultrasound for radiation therapy has been hindered by a few issues such as skin and target deformations due to probe pressure, optical tracking disabilities caused by irregular surfaces and inter-user variations. The purpose of this study was to overcome these barriers by using a patient-specific three-dimensional (3D) couplant pad (CP).
A patient skin mold was designed using a skin contour of simulation CT images and fabricated by a 3D printer. A CP was then casted by pouring gelatin solution into a container accommodating the mold. To validate the use of the CP in positioning accuracy and imaging quality, phantom tests were carried out in our ultrasound-based localization system and then daily ultrasound images of four patients were acquired with and without the CP before treatment.
In the phantom study, the use of CP increased a contrast-to-noise ratio from 2.4 to 4.0. The positioning accuracies in the US scans with and without the CP were less than 1 mm in all directions. In the patient study, the use of CP decreased the centroid offset of the target volume after target position alignment from 4.4 mm to 2.9 mm. One patient with a small volume of target showed a substantial increase in the inter-fractional target contour agreement (from 0.07 (poor agreement) to 0.31 (fair agreement) in Kappa values) by using the CP.
Our patient-specific 3D CP based on a 3D mold printing technique not only maintained the tracking accuracy but also reduced the inter-user variation, as well as that could potentially improve detectability of optical markers and target visibility for ultrasound image-guided radiotherapy.
Enke KA C, Saw CB, Zhen W, Thompson RB, Raman NV. Inter-Observer Variation in Prostate Localization Utilizing BAT. Int J Radiat Oncol Biol Phys. 2002;54(2):269. CrossRef
Lachaine M, Falco T. Intrafractional prostate motion management with the clarity autoscan system. Med Phys Int. 2013;1:72–80.
American College of Radiology. Acr practice parameter for the performance of a breast ultrasound examination, Amended 2014 (Resolution 39). Reston: ACR (American College of Radiology); 2014.
Yasukawa K, Kunisue T, Tsuta K, Shikinami Y, Kondo T. P6C-9 An Ultrasound Phantom with Long-Term Stability Using a New Biomimic Soft Gel Material. In: Ultrasonics Symposium, 2007 IEEE; 2007. p. 2501–2. CrossRef
Biller DS, Myer W. Ultrasound scanning of superficial structures using an ultrasound standoff pad. Vet Radiol. 1988;29:138–42. CrossRef
Yeh HC, Wolf BS. A simple portable water bath for superficial ultrasonography. Am J Roentgenol. 1978;130:275–8. CrossRef
Amin VR. Tissue characterization by ultrasound: a study of tissue-mimicking materials and quantitative ultrasonics, Retrospective Theses and Dissertations: Iowa State University; 1992. http://lib.dr.iastate.edu/.
Fleiss JL. Statistical methods for rates and proportions. 2nd ed. New York: Wiley. 1981. p. 38–46.
Mohler ER. Essentials of Vascular Laboratory Diagnosis. Hoboken: Wiley; 2008.
Boda-Heggemann J, Kohler FM, Kupper B, Wolff D, Wertz H, Mai S, Hesser J, Lohr F, Wenz F. Accuracy of ultrasound-based (BAT) prostate-repositioning: a three-dimensional on-line fiducial-based assessment with cone-beam computed tomography. Int J Radiat Oncol Biol Phys. 2008;70:1247–55. CrossRefPubMed
- A patient-specific three-dimensional couplant pad for ultrasound image-guided radiation therapy: a feasibility study
Ah Ram Chang
- BioMed Central
Neu im Fachgebiet Onkologie
Mail Icon II