Edwin HG Oei, Peter Pilotm, Gert-Jan Kleinrensink and Rolf M Bloem contributed equally to this work.
The authors declare that they have no competing interests.
MAR, MD. First author, main contributor of study design, experiments, data analysis and manuscript preparation. MIV MD Had major participation in study design, experiments and manuscript preparation. EHGO, MD PhD. Head of radiology department, important role in experiments and revision of manuscript. PP PhD. Senior research member, approvel of study design, important part of manuscript preparation and revision. GJK PhD. Head of department of Anatomy, essential for experiments on cadaver samples. RMB MD PhD. Head of department of Orthopaedic surgery, significant contribution to study design and manuscript revision. All authors read and approved the final manuscript.
Femoroacetabular impingement (FAI) is caused by an anatomic deviation of the acetabular rim or proximal femur, which causes chronic groin pain. Radiological identification of FAI can be challenging. Advances in imaging techniques with the use of computed tomography (CT) scan enable 3D simulation of FAI. We made an experimental cadaveric validation study to validate the 3D simulation imaging software.
The range of motion (ROM) of five cadaveric hips was measured using an electromagnetic tracking system (EMTS). Specific marked spots in the femur and pelvis were created as reproducible EMTS registration points. Reproducible motions were measured. Hips were subsequently imaged using high-resolution CT after introduction of artificial cam deformities. A proprietary software tool was used, Articulis (Clinical Graphics) to simulate the ROM during the presence and absence of the induced cam deformities.
According to the EMTS, 13 of the 30 measured ROM end-points were restricted by > 5° due to the induced cam deformities. Using Articulis, with the same 5° threshold, we correctly detected 12 of these 13 end point limitations and detected no false positives. The median error of the measured limitations was 1.9° (interquartile range 1.1° - 4.4°). The maximum absolute error was 5.4°.
The use of this dynamic simulation software to determine the presence of motion limiting deformities of the femoroacetabular is validated. The simulation software is able to non-invasively detect a reduction in achievable ROM, caused by a cam type deformity.
Byrd J, Jones K. Hip arthroscopy in athletes: 10-year follow-up. Am J Sports Med. 2009;11(37):2140–3. CrossRef
Hack K, Primio G, Rakhra K. Prevalence of cam-type femoroacetabular impingement morphology in asymptomatic volunteers. J Bone Joint Surg. 2010;14(92):2436–44. CrossRef
Aprato A.; Jayasekera N.; Viaal RN. Revision hip arthroscopic surgery: outcome at three years. Knee Surg. Sports Traumatol. Arthrosc. 2013; Jan 18, doi:10.1007/s001670-013-2373-7
Harris JD, Erickson BJ, Bush-Joseph CA, Nho SJ. Treatment of femoroacetabular impingement: a systematic review. Curr Rev Muscoloskelet Med. 2013;6:207–18. CrossRef
Lepage-Saucier M, Thiéry C, Larbi A, Lecouvet FE, Vand Berg BC, Omoumi P. Femoroacetabular impingement: normal values of the quantitative morphometric parameters in asymptomatic hips. Eur Radiol. 2014;7:1707–14. CrossRef
Krekel PR, Botha CP, Vastart ER, de Bruik RW, Rozing PM, Post FH. “Interactive simulation and comparative visualisation of the bone-determinded range of motion of the human shoulder”. Proc Simulation Visualization. 2006;3:275–88.
Beck M, Kalhor M, Leunig M, Ganz R. Hip morphology influences the pattern of damage to the acetabular cartilage: femoroacetabular impingement as a cause of early osteoarthritis of the hip. J Boint Joint Surg (Br). 2005;87:1012–8. CrossRef
Becce F, Ben Salah Y, Verdun FR, Vande Berg BC, Lecouvet FE, Leuli R, et al. Computed tomography of the cervical spine: comparison of image quality between a standard-dose and a low-dose protocol using filtered back-projection and iterative reconstruction. Skeletal Radiol. 2013;42(7):937–45. CrossRefPubMed
- A quantitative non-invasive assessment of femoroacetabular impingement with CT-based dynamic simulation - cadaveric validation study
Maarten A Röling
Monique I Visser
Edwin HG Oei
Rolf M Bloem
- BioMed Central
Neu im Fachgebiet Orthopädie und Unfallchirurgie
Mail Icon II