Correction to: European Radiology
The original version of this article, published on 10 February 2019, unfortunately contained a mistake. The axes of the graphs in Fig. 3 are incorrect. The correct figure is given below. Therefore, the last two sentences in “Results”, section “Noise”, should read: “The peak frequency of the HR and SHR was 0.21 lp/mm. For the NR mode and the MDCT, the peak frequencies were 0.17 lp/mm and 0.21 lp/mm, respectively.”
The X-axis of the graphs in Fig. B2 (Addendum B) has the same issue. The correct figure is given below. These changes do not affect any conclusion drawn from the results.
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons licence, and indicate if changes were made. The images or other third party material in this article are included in the article's Creative Commons licence, unless indicated otherwise in a credit line to the material. If material is not included in the article's Creative Commons licence and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this licence, visit http://creativecommons.org/licenses/by/4.0/.
- Correction to: Physical evaluation of an ultra-high-resolution CT scanner
Luuk J. Oostveen
Kirsten L. Boedeker
Frank de Lange
- Springer Berlin Heidelberg
Print ISSN: 0938-7994
Elektronische ISSN: 1432-1084
Neu im Fachgebiet Radiologie
Meistgelesene Bücher aus der Radiologie
Mail Icon II