The online version of this article (doi:10.1186/1757-1146-7-5) contains supplementary material, which is available to authorized users.
The authors declare that they have no competing interests.
AKT conducted the main research, design, and testing of the system. RJA provided support on the clinical aspects of the study and was the main party in securing funding. ET and JS supplied advice, as well as coding tools and testing tools for the computational side of the project. WW carried out the relevant statistical analysis. All authors read and approved the final manuscript.
The foot provides a crucial contribution to the balance and stability of the musculoskeletal system, and accurate foot measurements are important in applications such as designing custom insoles/footwear. With better understanding of the dynamic behavior of the foot, dynamic foot reconstruction techniques are surfacing as useful ways to properly measure the shape of the foot. This paper presents a novel design and implementation of a structured-light prototype system providing dense three dimensional (3D) measurements of the foot in motion. The input to the system is a video sequence of a foot during a single step; the output is a 3D reconstruction of the plantar surface of the foot for each frame of the input.
Engineering and clinical tests were carried out to test the accuracy and repeatability of the system. Accuracy experiments involved imaging a planar surface from different orientations and elevations and measuring the fitting errors of the data to a plane. Repeatability experiments were done using reconstructions from 27 different subjects, where for each one both right and left feet were reconstructed in static and dynamic conditions over two different days.
The static accuracy of the system was found to be 0.3 mm with planar test objects. In tests with real feet, the system proved repeatable, with reconstruction differences between trials one week apart averaging 2.4 mm (static case) and 2.8 mm (dynamic case).
The results obtained in the experiments show positive accuracy and repeatability results when compared to current literature. The design also shows to be superior to the systems available in the literature in several factors. Further studies need to be done to quantify the reliability of the system in clinical environments.
Additional file 1: Supplementary material.(PDF 657 KB)13047_2013_589_MOESM1_ESM.pdf
Authors’ original file for figure 113047_2013_589_MOESM2_ESM.pdf
Authors’ original file for figure 213047_2013_589_MOESM3_ESM.pdf
Authors’ original file for figure 313047_2013_589_MOESM4_ESM.pdf
Authors’ original file for figure 413047_2013_589_MOESM5_ESM.pdf
Authors’ original file for figure 513047_2013_589_MOESM6_ESM.pdf
Authors’ original file for figure 613047_2013_589_MOESM7_ESM.pdf
Authors’ original file for figure 713047_2013_589_MOESM8_ESM.pdf
Authors’ original file for figure 813047_2013_589_MOESM9_ESM.pdf
Authors’ original file for figure 913047_2013_589_MOESM10_ESM.pdf
Authors’ original file for figure 1013047_2013_589_MOESM11_ESM.pdf
Authors’ original file for figure 1113047_2013_589_MOESM12_ESM.pdf
Abboud RJ: Relevant foot biomechanics. Curr Orthop. 2002, 16: 165-179. 10.1054/cuor.2002.0268. CrossRef
Payne C: Cost benefit comparison of plaster casts and optical scans of the foot for the manufacture of foot orthoses. Australas J Podiatric Med. 2007, 41: 29-31.
Shuh-Ping S, Yi-Jiun C: Foot 3D image capture modules under non-weight bearing state. J Biomech. 2007, 40: S52- CrossRef
Martedi S, Saito H: Shape measurement system of foot sole surface from flatbed scanner image. MVA2009 IAPR Conference on Machine Vision Applications, 20–22nd June 2009. 2009, Keio, Tokyo, Japan: MVA,
Amstutz E, Teshima T, Kimura M, Mochimaru M, Saito H: PCA based 3D shape reconstruction of human foot using multiple viewpoint cameras. Int J Automation Comput. 2008, 5008: 161-170.
Telfer S, Woodburn J, et al: The use of 3D surface scanning for the measurement and assessment of the human foot. J Foot Ankle Res. 2010, 5: 19- CrossRef
Trucco E, Verri A: Introductory Techniques for 3-D Computer Vision. 1998, Upper Saddle River, London: Prentice Hall
Coudert T, Vacher P, Smits C, Van Der Zande M: A method to obtain 3D foot shape deformation during the gait cycle. 9th International Symposium on the 3D analysis of Human Movement, 28–30th June 2006 Valenciennes, France. 2006, Pennsylvania: ISB,
Jezersek M, Mozina J: High-speed measurement of foot shape based on multiple-laser-plane triangulation. Opt Eng. 2009, 48 (11): 113604-113604. 10.1117/1.3265522. CrossRef
Salvi J, Fernandez S, Pribanic T, Llado X: A State of the art in structured light patterns for surface profilometry. Pattern Recognit. 2010, 43 (8): 2666-2680. 10.1016/j.patcog.2010.03.004. CrossRef
Jezersek M, Novak B, Mozina J: Three-dimensional laser based measurement of human foot during walking. Footwear Sci. 2011, 3 (sup1): S81-S83. 10.1080/19424280.2011.575867. CrossRef
Schmeltzpfenning T, Plank C, Krauss I, Aswendt P, Grau S: Dynamic foot scanning: a new approach for measurement of the human foot shape while walking. Footwear Sci. 2009, 1 (sup1): S28-S30. 10.1080/19424280902977111. CrossRef
Schmeltzpfenning T, Plank C, Fritz B, Aswendt P, Grau S: 3D dynamic behaviour of foot structure may provide additional information for last design. Footwear Sci. 2011, 3 (sup1): S147-S148. 10.1080/19424280.2011.575866. CrossRef
Mochimaru M, Kouchi M: 4D measurement and analysis of plantar deformation during walking and running. Footwear Sci. 2011, 3 (sup1): S109-S112. 10.1080/19424280.2011.575878. CrossRef
Yoshida Y, Saito S, Aoki Y, Kouchi M, Mochimaru M: Shape completion and modeling of 3D foot shape while walking. International Symposium on Optomechatronic Technologies (ISOT). 2012, New York: IEEE, 1-3.
Liu S, Cui Y, Sanchez S, Stricker D: Foot scanning and deformation estimation using time-of-flight cameras. Footwear Sci. 2011, 3 (sup1): S98-S99. 10.1080/19424280.2011.575862. CrossRef
Sturmer M, Seiler C, Becker G, Hornegger J: Alignment of multiple time-of-flight 3D cameras for reconstruction of walking feet. Proceedings of ISB XXIII: 3–7th July 2011 Brussels, Belgium. 2011, Pennsylvania: ISB,
Blenkinsopp R, Harland A, Price D, Lucas T, Roberts J: A method to measure dynamic dorsal foot surface shape and deformation during linear running using digital image correlation. Procedia Eng. 2012, 34 (0): 266-271. CrossRef
Pages J, Salvi J, Collewet C, Forest J: Optimised de Bruijn patterns for one-shot shape acquisition. Image Vis Comput. 2005, 23: 707-720. 10.1016/j.imavis.2005.05.007. CrossRef
Thabet AK, Trucco E, Salvi J, Wang W, Abboud RJ: A dynamic 3D foot reconstruction system. Annual International Conference of the IEEE Engineering in Medicine and Biology Society, 30 August - 3 September 2011 Boston, USA. 2011, New York: IEEE,
Besl PJ, McKay ND: A method for registration of 3-D shapes. IEEE Trans Pattern Anal Mach Intell. 1992, 14: 239-256. 10.1109/34.121791. CrossRef
Trucco E, Fusiello A, Roberto V: Robust motion and correspondences of noisy 3-D point sets with missing data. Pattern Recognit Lett. 1999, 20: 889-898. 10.1016/S0167-8655(99)00055-0. CrossRef
Salvi J, Armangue X, Batlle J: A comparative review of camera calibrating methods with accuracy evaluation. Pattern Recognit. 2002, 35: 1617-1635. 10.1016/S0031-3203(01)00126-1. CrossRef
- Dynamic 3D shape of the plantar surface of the foot using coded structured light: a technical report
Ali K Thabet
Rami J Abboud
- BioMed Central
Neu im Fachgebiet Orthopädie und Unfallchirurgie
Mail Icon II