MR imaging of neonates is difficult for many reasons and a major factor is safe transport to the MR facilities. In this article we describe the use of a small, investigational 3-T MR customised for brain imaging and sited on a neonatal unit of a tertiary centre in the UK, which is in contrast to a 300-m journey to the whole-body MR scanner used at present for clinical cases.
We describe our methods for preparing babies for safe transport and scanning on an investigational 3-T MR scanner on a neonatal unit and the development of appropriate MR sequences. The MR scanner does not have CE marking at present so this early development work was undertaken on normal neonates whose parents consented to a research examination.
Fifty-two babies were scanned and there were no serious adverse events. The MR examinations were considered to be diagnostically evaluable in all 52 cases and in 90% the imaging was considered to be at least as good as the quality obtained on the 1.5-T scanner currently used for clinical cases.
We have shown that this investigational 3-T MR scanner can be used safely on a neonatal unit and we have refined the MR sequences to a point that they are clinically usable.
• Access to neonatal MR imaging is limited.
• We describe an investigational 3-T MR scanner site on a neonatal unit.
• The scanner produces images suitable for clinical practice.
Mosaic Science (2016) Tiny brain scanner slashes newborns’ hospital commute. Available via https://mosaicscience.com/story/baby-mri-premature-brain-scan-tech. Accessed 22 Jan 2018
International Electrotechnical Commission (2008) Medical electrical equipment – Particular requirements for the safety of MR equipment for medical diagnosis IEC 60601-2-33 ed 2.2 (Geneva IEC) with amendments 1 and 2
Whitby EH, Griffiths PD, Lonneker-Lammers T et al (2004) Ultrafast magnetic resonance imaging of the neonate in a magnetic resonance-compatible incubator with a built-in coil. Pediatrics 113(2):150–152 CrossRef
Paley MN, Hart AR, Lait M, Griffiths PD (2012) An MR-compatible neonatal incubator. Br J Radiol 85(1015):952–983 CrossRef
Merchant N, Groves A, Larkman DJ et al (2009) A patient care system for early 3.0 Tesla magnetic resonance imaging of very low birth weight infants. Early Hum Dev 85(12):779–783 CrossRef
Hughes EJ, Winchman T, Padormo F et al (2017) A dedicated neonatal brain imaging system. Magn Reson Med 78(2):794–804 CrossRef
Whitby EH, Griffiths PD, Rutter S et al (2004) Frequency and natural history of subdural haemorrhages in babies and relation to obstetric factors. Lancet 363(9412):846–851 CrossRef
Lother S, Schiff SJ, Neuberger T, Jakob PM, Fidler F (2016) Design of a mobile, homogeneous, and efficient electromagnet with a large field of view for neonatal low-field MRI. MAGMA 29(4):691–698 CrossRef
Aspect Imaging (2017) A new approach to MRI in the NICU. Available via https://www.neonatalmri.com. Accessed 22 Jan 2018
Tkach JA, Merhar SL, Kline-Fath BM et al (2014) MRI in the neonatal ICU: initial experience using a small-footprint 1.5-T system. Am J Roentgenol 202(1):95–105 CrossRef
Merhar SL, Tkach JA, Woods JC et al (2017) Neonatal imaging using an on-site small footprint MR scanner. Pediatr Radiol 47(8):1001–1011 CrossRef
U.S. Food & Drugs Administration (2014) Criteria for significant risk investigation of magnetic resonance resonance diagnostic devices. Available via http://www.fda.gov/MedicalDevices/DeviceRegulationandGuidance.htm. Accessed 22 Jan 2018
Cawley P, Few K, Greenwood R et al (2016) Does magnetic resonance brain scanning at 3.0 Tesla pose a hyperthermic challenge to term neonates? J Pediatrics 175:228-30.e1 CrossRef
- Initial experience of an investigational 3T MR scanner designed for use on neonatal wards
Paul D. Griffiths
Daniel J. A. Connolly
Anthony R. Hart
Martyn N. J. Paley
- Springer Berlin Heidelberg
Neu im Fachgebiet Radiologie
Meistgelesene Bücher aus der Radiologie
Mail Icon II