The present study evaluated the temporal release of Co Cr, Mn, and Ni from the components of a typical orthodontic appliance during simulated orthodontic treatment.
Several commercially available types of bands, brackets, and wires were exposed to an artificial saliva solution for at least 44 days and the metals released were quantified in regular intervals using inductively coupled plasma quadrupole mass spectrometry (ICP-MS, Elan DRC+, Perkin Elmer, USA). Corrosion products encountered on some products were investigated by a scanning electron microscope equipped with an energy dispersive X-ray microanalyzer (EDX).
Bands released the largest quantities of Co, Cr, Mn, and Ni, followed by brackets and wires. Three different temporal metal release profiles were observed: (1) constant, though not necessarily linear release, (2) saturation (metal release stopped after a certain time), and (3) an intermediate release profile that showed signs of saturation without reaching saturation. These temporal metal liberation profiles were found to be strongly dependent on the individual test pieces. The corrosion products which developed on some of the bands after a 6-month immersion in artificial saliva and the different metal release profiles of the investigated bands were traced back to different attachments welded onto the bands.
The use of constant release rates will clearly underestimate metal intake by the patient during the first couple of days and overestimate exposure during the remainder of the treatment which is usually several months long. While our data are consistent with heavy metal release by orthodontic materials at levels well below typical dietary intake, we nevertheless recommend the use of titanium brackets and replacement of the band with a tube in cases of severe Ni or Cr allergy.
Arndt M, Brück A, Scully T, Jäger A, Bourauel C (2005) Nickel ion release from orthodontic NiTi wires under simulation of realistic in situ conditions. J Mater Sci 40(14):3659–3667 CrossRef
Barrett RD, Bishara SE, Quinn JK (1993) Biodegradation of orthodontic appliances. Part I. Biodegradation of nickel and chromium in vitro. Am J Orthod Dentofac Orthop 103(1):8–14 CrossRef
Buehler WJ, Gilfrich JV, Wiley RC (1963) Effect of low temperature phase changes on the mechanical properties of alloys near composition TiNi. J Appl Phys 34(5):1475–1477 CrossRef
Eliades T, Athanasiou AE (2002) In vivo aging of orthodontic alloys: implications for corrosion potential, nickel release, and biocompatibility. Angle Orthod 72(3):222–237 PubMed
Flyvholm MA, Nielsen G, Andersen A (1984) Nickel content of food and estimation of dietary intake. Z Lebensm Unters Forch 179(6):427–431 CrossRef
Greppi AL, Smith DC, Woodside DG (1989) Nickel hypersensitivity reactions in orthodontic patients. A literature review. Univ Tor Dent J 3(1):11–14 PubMed
Grimsdottir MR, Gjerdet NR, Hensten-Pettersen A (1992) Composition and in vitro corrosion of orthodontic appliances. Am J Orthod Dentofac Orthop 101(6):525–532 CrossRef
Huang TH, Yen CC, Kao CT (2001) Comparison of ion release from new and recycled orthodontic brackets. Am J Orthod Dentofac Orthop 120(1):68–75 CrossRef
Kim H, Johnson JW (1999) Corrosion of stainless steel, nickel–titanium, coated nickel–titanium, and titanium orthodontic wires. Angle Orthod 69:39–44 PubMed
Kerosuo H, Moeb G, Hensten-Pettersen A (1997) Salivary nickel and chromium in subjects with different types of fixed orthodontic appliances. Am J Orthod Dentofac Orthoped 111(6):595–598 CrossRef
Moffa JP (1984) Biocompatibility of nickel based dental alloys. J Calif Dent Assoc 12(10):45–51
Ortiz AJ, Fernández E, Vicente A, Calvo JL, Ortiz C (2011) Metallic ions released from stainless steel, nickel-free, and titanium orthodontic alloys: toxicity and DNA damage. Am J Orthod Dentofac Orthop 140(3):e115–e122 CrossRef
Petoumenou E, Arndt M, Keilig L, Reimann S, Hoederath H, Eliades T, Jäger A, Bourauel C (2009) Nickel concentration in the saliva of patients with nickel–titanium orthodontic appliances. Am J Orthod Dentofac Orthop 135(1):59–65 CrossRef
Pulikkottil VJ, Chidambaram S, Bejoy PU, Femin PK, Paul P, Rishad M (2016) Corrosion resistance of stainless steel, nickel-titanium, titanium molybdenum alloy, and ion-implanted titanium molybdenum alloy archwires in acidic fluoride-containing artificial saliva: an in vitro study. J Pharm Bioallied Sci 8(Suppl 1):96–99
Reaxys Database (Beilstein, Gmelin) (2013) Reed Elsevier Properties SA
Rubin R, Strayer DS, Emanuel R (2008). Rubin’s Pathology:Clinicopathologic Foundations of Medicine 6th ed
Standardization Ecf (2011) Reference test method for release of nickel from all post assemblies which are inserted into pierced parts of the human body and articles intended to come into direct and prolonged contact with the skin: Austrian Standards plus GmbH.Ref.nr.EN 1811:1998D; http://www.ris.bka.gv.at
Tanner SD, Baranov VI (1999) Theory, design, and operation of a dynamic reaction cell for ICP-MS. At Spectrosc 20(2):45–52
Tanner SD, Baranov VI, Vollkopf U (2000) A dynamic reaction cell for inductively coupled plasma mass spectrometry (ICP-DRC-MS). Part III. Optimization and analytical performance. J Anal At Spectrom 15(9):1261–1269 CrossRef
- Metal release profiles of orthodontic bands, brackets, and wires: an in vitro study
- Springer Medizin
Journal of Orofacial Orthopedics / Fortschritte der Kieferorthopädie
Official Journal of the German Orthodontic Society / Offizielle Zeitschrift der Deutschen Gesellschaft für Kieferorthopädie
Print ISSN: 1434-5293
Elektronische ISSN: 1615-6714
Neu im Fachgebiet Zahnmedizin
Mail Icon II