To assess the precision and agreement of measurements of higher order aberrations (HOAs) obtained with a ray tracing aberrometer (iTrace) and a Hartmann-Shack aberrometer (Topcon KR-1 W).
Prospective evaluation of the diagnostic test. Data from the right eyes of 92 normal subjects obtained using the two devices were included in this study. Two observers performed 3 consecutive scans to determine the intraobserver repeatability and interobserver reproducibility. About one week later, one observer performed an additional 3 consecutive scans to obtain the intersession reproducibility. The within-subject standard deviation (Sw), test-retest repeatability (TRT) and intraclass correlation coefficient (ICC) were used to assess the precision, while Bland-Altman plots were performed to assess the agreement.
For intraobserver repeatability of the ocular, corneal and internal HOAs, Topcon KR-1 W showed a 2.77Sw of 0.079 μm or less and ICCs of 0.761 or more; and iTrace showed a 2.77Sw of 0.105 μm or less and ICCs of 0.805 or more. The ICCs of the internal HOAs of interobserver reproducibility were less than 0.75 except for spherical aberration (SA) (0.862), and interobserver reproducibility of the counterpart showed similar but lower results. For the ocular, corneal and internal HOA measurements, statistically significant differences existed between the Topcon KR-1 W and iTrace (all P < 0.05). No significant differences were observed in the ocular SA and internal coma.
The ray tracing and Hartmann-Shack method aberrometers provided excellent repeatability but less reliable reproducibility in the measurement of HOAs (except for SA). The two aberrometers should not be interchangeable in clinical application because of the significant differences in HOA measurements between them.
Cervino A, Hosking SL, Montes-Mico R, Bates K. Clinical ocular wavefront analyzers. J Refract Surg. 2007;23(6):603–16. PubMed
Montes-Mico R, Caliz A, Alio JL. Wavefront analysis of higher order aberrations in dry eye patients. J Refract Surg. 2004;20(3):243–7. PubMed
Gobbe M, Reinstein DZ, Archer TJ. LASIK-induced aberrations: comparing corneal and whole-eye measurements. Optometry & Vision Science. 2015;92(4):447–55. CrossRef
Piñero DP, Sánchez-Pérez PJ, Alió JL. Repeatability of measurements obtained with a ray tracing aberrometer. Optometry & Vision Science. 2011;88(9):1099–105. CrossRef
Hua Y, Xu Z, Qiu W, Wu Q. Precision (repeatability and reproducibility) and agreement of corneal power measurements obtained by Topcon KR-1W and iTrace. PLoS One. 2016;11(1):e0147086.
Institution BS. Accuracy (trueness and precision) of measurement methods and results - part 1: general principles and definitions. 1994.17. Bland a: measurement error and correlation coefficients. BMJ. 1996;313:41–2. CrossRef
Pollack K. Diurnal fluctuation of higher order ocular aberrations: correlation with intraocular pressure and corneal thickness. J Refract Surg. 2004;20(3):236. PubMed
Miranda MA, O'Donnell C, Radhakrishnan H. Repeatability of corneal and ocular aberration measurements and changes in aberrations over one week. Clinical & experimental optometry. 2009;92(3):253–66. CrossRef
Knapp S, Awwad ST, Ghali C, JP MC. Ocular aberrations measured by the Fourier-based WaveScan and Zernike-based LADARWave Hartmann-shack aberrometers. J Refract Surg. 2009;25(2):201. PubMed
Charman WN. Wavefront technology: past, present and future. Contact lens & anterior eye : the journal of the British Contact Lens Association. 2005;28(2):75–92. CrossRef
Cheng X, Himebaugh NL, Kollbaum PS, Thibos LN, Bradley A. Validation of a clinical shack-Hartmann aberrometer. Optometry and vision science : official publication of the American Academy of Optometry. 2003;80(8):587–95. CrossRef
Cervino A, Hosking SL, Rai GK, Naroo SA, Gilmartin B. Wavefront analyzers induce instrument myopia. J Refract Surg. 2006;22(8):795–803. PubMed
MacRae S, Fujieda M. Slit skiascopic-guided ablation using the Nidek laser. J Refract Surg. 2000;16(5):S576–80. PubMed
- Precision and agreement of higher order aberrations measured with ray tracing and Hartmann-Shack aberrometers
- BioMed Central
Neu im Fachgebiet Augenheilkunde