The BD MAX™ Enteric Bacterial Panel (BDM-EBP) is designed and FDA-cleared to detect Salmonella, Shigella, Campylobacter, and Shiga toxin genes stx1/2 from stool samples. However, rectal swabs, which are not FDA-cleared for clinical testing with the BDM-EBP, are common specimens received from pediatric patients for enteric pathogen testing. The purpose of this study was to evaluate the ability of the BDM-EBP to detect stool pathogens from rectal swabs.
Routine cultures, Shiga toxin testing, and molecular testing with BDM-EBP were performed on 272 sequential rectal swabs collected from August 2015 to December 2015. Discrepant test results were resolved using Verigene® Enteric Pathogens Nucleic Acid Test (EP). 36 challenge samples (13 Salmonella spp., 3 Shigella spp., 10 Campylobacter spp., and 10 Shiga toxin positive Escherichia coli) were tested using reference strains (American Type Culture Collection) and previous patient isolates diluted to103-104 cfu/ml in saline then added to Sample Buffer Tube (SBT) with negative stool matrix delivered via a swab. Limit of detection testing was performed by serial 10 fold dilutions in saline then added to SBT with negative stool matrix provided via a swab.
A total of 272 rectal swab specimens were evaluated and 89 were positive by culture and/or MAX EBP. All discrepant results were BDM-EBP positive and culture negative. 21 of 31 (68%) of the apparent false positive BDM-EBP discrepant results resolved as positive with Nanosphere’s Verigene® EP. After resolution of the discordant results, the Positive Percent Agreement (PPA) and Negative Percent Agreement (NPA) are as follows for each target: Salmonella (n = 4) 100%, PPA and 100%, NPA; Shigella (n = 79) 100%, PPA and 95.3%, NPA; Campylobacter (n = 4) 100%, PPA and 99.6%, NPA; and Shiga toxin producing organisms (n = 2) 100%, PPA and 100%, NPA. 8.8% of the patient samples did not initially yield a result on the BDM-System. Upon repeat, half of the problematic samples resolved, and 4.4% of the total specimen tested did not yield a result. All organisms in the challenge samples were detected. Limits of detection for BDM-EBP testing of rectal swabs were as follows (in cfu/ml in SBT): Salmonella-1.44 × 102; Shigella-5.10 × 100; Campylobacter-1.51 × 101; and Shiga Toxin-1.13 ×103.
Rectal swabs are acceptable samples for detecting Salmonella, Shigella, Campylobacter, and Shiga toxin using BDM-EBP.
WHO/UNICEF. Ending preventable child deaths from pneumoniae and diarrhea by 2025: The integrated Global Action Plan for Pneumoniae and Diarrhoeae (GAPPD). New York: The United Nations Children’s Fund (UNICEF)/World Health Organization (WHO); 2013.
Johansson EW, Wardlaw T, Binkin N, Brocklehurst C, Dooley T. Diarrhoea: why children are still dying and what can be done, (UNICEF). New York: The United Nations Children’s Fund (UNICEF)/World Health Organization (WHO); 2009.
Harrington SM, Buchan BW, Doern C, Fader R, Ferraro MJ, Pillai DR, Rychert J, Doyle L, Lainesse A, Karchmer T, Mortensen JE. Multicenter Evaluation of the BD Max™ Enteric Bacterial Panel PCR Assay for Rapid Detection of Salmonella spp., Shigella spp., Campylobacter spp. ( C. jejuni and C. coli), and Shiga Toxin 1 and 2 Genes. J Clin Microbiol. 2015;53:1639–46. CrossRefPubMedPubMedCentral
Garcia LS, Isenberg HD, Clinical Microbiology Procedures Handbook . 3 rd ed. 2010. Washington, DC: ASM Press, pp. 2.1.14, 184.108.40.206, 220.127.116.11.
Jorgensen JH, et al., Manual of Clinical Microbiology. 11 th ed. 2015, Washington, DC: ASM Press, pp. 271, 275, 276.
- Utilizing BD MAX™ Enteric Bacterial Panel to Detect Stool Pathogens from Rectal Swabs
Eleanor A. Powell
Joel E. Mortensen
- BioMed Central