Here we present how the CYBERLEGs Beta-Prosthesis was modified with a new control system to participate in the Powered Leg Prosthesis event, and to report on our experience at the CYBATHLON 2016 which was held in Zurich, Switzerland in October 2016. The prosthesis has two active degrees of freedom which assist the user with extra joint power at the knee and ankle to complete tasks. The CYBATHLON is a championship for people with disabilities competing in six disciplines, using advanced assistive devices. Tasks for CYBATHLON 2016 were chosen to reflect everyday normal task such as sitting and standing from a chair, obstacle avoidance, stepping stones, slope walking and descent, and stair climbing and descent.
The control schemata were presented along with the description of each of the six tasks. The participant of the competition, the pilot, ran through each of the trials under lab conditions and representative behaviors were recorded.
The VUB CYBERLEGs prosthesis was able to accomplish, to some degree, five of the six tasks and here the torque and angle behaviors of the device while accomplishing these tasks are presented. The relatively simple control methods were able to provide assistive torque during many of the events, particularly sit to stand and stair climbing. For example, the prosthesis was able to consistently provide over 30 Nm in arresting knee torque in the sitting task, and over 20 Nm while standing. Peak torque of the device was not sufficient for unassisted stair climbing, but was able to provide around 60 Nm of assistance in both ascent and descent. Use of the passive behaviors of the device were shown to be able to trigger state machine events reliably for certain tasks.
Although the performance of the CYBERLEGs prosthesis during CYBATHLON 2016 did not compare to the other top of the market designs with regards to speed, the device performed all of the tasks that were deemed possible by the start of the competition. Moreover, the Pilot was able to accomplish tasks in ways the Pilot’s personal microcontrolled prosthesis could not, with limited powered prosthesis training. Future studies will focus on decreasing weight, increasing reliability, incorporating better control, and increasing the velocity of the device. This is only a case study and actual benefits to clinical outcomes are not yet understood and need to be further investigated. This competition was a unique experience to illuminate problems that future versions of the device will be able to solve.
Geeroms J, Flynn L, Jimenez-Fabian R, Vanderborght B, Lefeber D. Energetic analysis and optimization of a maccepa actuator in an ankle prosthesis. Auton Robot. 2017. doi: 10.1007/s10514-017-9641-1.
Flynn L, Geeroms J, Jimenez-Fabian R, Vanderborght B, Lefeber D. Cyberlegs beta-prosthesis active knee system. In: 2015 IEEE International Conference on Rehabilitation Robotics (ICORR): 2015. p. 410–5. doi: 10.1109/ICORR.2015.7281234.
Winter DA. Biomechanics and Motor Control of Human Movement, 4th ed.United States of America: Wiley; 2009, p. 384. CrossRef
Jimenez-Fabian R, Flynn L, Geeroms J, Vitiello N, Vanderborght B, Lefeber D. Sliding-Bar MACCEPA for a Powered Ankle Prosthesis. J Mech Robot. 2015; 7(March):1–2. doi: 10.1115/1.4029439.
Holgate MA, Sugar TG, Alexander WB. A Novel Control Algorithm for Wearable Robotics using Phase Plane Invariants. IEEE Int Conf Robot Autom. 2009; May:3845–50.
Grosu V, Guerrero CR, Brackx B, Grosu S, Vanderborght B, Lefeber D. Instrumenting complex exoskeletons for improved human-robot interaction. IEEE Instrum Meas Mag. 2015; 18(5):5–10. doi: 10.1109/MIM.2015.7271219.
Sup F, Varol HA, Mitchell J, Withrow TJ, Goldfarb M. Preliminary Evaluations of a Self-Contained Anthropomorphic Transfemoral Prosthesis. IEEE/ASME Trans Mechatron Joint Publ IEEE Ind Electron Soc ASME Dyn Syst Control Div. 2009; 14(6):667–76. doi: 10.1109/TMECH.2009.2032688 .
Ledoux ED, Lawson BE, Shultz AH, Bartlett HL, Goldfarb M. Metabolics of stair ascent with a powered transfemoral prosthesis. In: 2015 37th Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBC): 2015. p. 07–310. doi: 10.1109/EMBC.2015.7319589.
Varol HA, Sup F, Goldfarb M. Powered Sit-to-Stand and Assistive Stand-to-Sit Framework for a Powered Transfemoral Prosthesis. In: Proceedings of the 2009 IEEE International Conference on Rehabilitation Robotics, Kyoto, Japan: 2009. p. 645–51. doi: 10.1109/ICORR.2009.5209582.
Wolf EJ, Everding VQ, Linberg AL, Schnall BL, Czerniecki M, Gambel JM. Assessment of transfemoral amputees using C-Leg and Power Knee for ascending and descending inclines and steps. JRRD. 2012; 49(6):831–42. CrossRef
Lay AN, Hass CJ, Gregor RJ. The effects of sloped surfaces on locomotion: A kinematic and kinetic analysis. J Biomech. 2006; 39:1621–8. doi: 10.1016/j.jbiomech.2005.05.005.
- VUB-CYBERLEGs CYBATHLON 2016 Beta-Prosthesis: case study in control of an active two degree of freedom transfemoral prosthesis
Louis L. Flynn
Tom van der Hoeven
- BioMed Central
Neu in den Fachgebieten Neurologie und Psychiatrie
Meistgelesene Bücher in der Neurologie & Psychiatrie