John W Lockhart and John D Breckenridge contributed equally to this work.
The authors declare that they have no competing interests.
All authors participated in the study conception, design and coordination. ADH and JWL collected study data. ADH, JWL and JDB performed the data analysis. ADH wrote the first draft of the paper with input of all authors. All authors read and approved the final manuscript.
We undertook the current study to assess whether an accelerometer-based physical activity monitor, the SenseWear Mini Armband (SMA), could be used to provide data on static arm elevation, and to assess the agreement between static arm elevation measures obtained using SMA-derived data and those obtained with a universal goniometer.
Using a universal goniometer, healthy adult subjects (n = 25, age 30 ± 9 years) had each of right and left arms positioned in a series of set positions between arm-by-side and maximal active arm flexion (anteversion), and arm-by-side and maximal active arm abduction. Subjects wore the SMA throughout positioning, and SMA accelerometer data was used to retrospectively calculate/derive arm elevation angle using a manufacturer-provided algorithm. The Bland-Altman method was used to assess agreement between goniometer-set and SMA-derived arm elevation angles.
There were significant differences between goniometer-set and SMA-derived arm elevation angles for elevation angles ≤ 30 degrees and ≥ 90 degrees (p < 0.05). Bland-Altman plots showed that the greater the angle of elevation, the greater the mean difference between goniometer-set and SMA-derived elevation angles. Adjustment of the manufacturer-provided algorithm for deriving arm elevation angle corrected for this systematic difference, and resulted in 95% limits of agreement ± 12 degrees (flexion) and ± 13 degrees (abduction) across the full range of arm elevation.
The SMA can be used to record data allowing derivation of static arm elevation angle in the upright position, 95% limits of agreement with the universal goniometer being similar to those reported for digital inclinometers and gyroscopes. Physiotherapists looking for innovative methods of recording upper limb range of motion should consider the potential of accelerometer-based physical activity monitors such as the SMA.
Carey MA, Laird DE, Murray KA, Stevenson JR. Reliability, validity, and clinical usability of a digital goniometer. Work. 2010;36:55–66. PubMed
Reese NB, Bandy WD. Measurement of range of motion and muscle length: background, history, and basic principles. In: Falk K, editor. Joint range of motion and muscle length testing. 2nd ed. St Louis: Saunders Elsevier; 2010. p. 3–30.
Reese NB, Bandy WD. Reliability and validity of measurements of range of motion and muscle length testing of the upper extremit. In: Falk K, editor. Joint range of motion and muscle length testing. 2nd ed. St Louis: Saunders Elsevier; 2010. p. 165–82.
Kennedy CA, Beaton DE, Solway S, McConnell S, Bombardier C. Disabilities of the Arm, Shoulder and Hand (DASH). The DASH and QuickDASH outcome measure user’s manual. 3rd ed. Toronto: Institute for Work & Health; 2011.
Lugo K. Bodymedia Inc. 2014, Personal communication.
Rettig O, Krautwurst B, Wolf SI. Ex-/internal rotation during bilateral shoulder ab-/adduction and ante-/retroversion movements described as conjunct and adjunct rotation. Rettig O, UniversitatsKlinikum Heidelberg.2014, personal communication.
Amasay T, Zodrow K, Kincl L, Hess J, Karduna A. Validation of tri-axial accelerometer for the calculation of elevation angles. Int J Ind Ergon. 2009;39:783–9. CrossRef
- Can a physical activity monitor provide a valid measure of arm elevation angle? A study to assess agreement between the SenseWear Mini Armband and the universal goniometer
Andrew D Hirschhorn
John W Lockhart
John D Breckenridge
- BioMed Central
Neu im Fachgebiet Orthopädie und Unfallchirurgie
Mail Icon II