The online version of this article (doi:10.1186/1754-9493-8-3) contains supplementary material, which is available to authorized users.
Thomas Freude, Steffen Schröter contributed equally to this work.
The authors declare that they have no competing interests. None of the authors has any financial interests with regard to the companies whose products are described in this paper, i.e. DePuy-Synthes, Zuchwil, Switzerland.
All authors have contributed significantly to the different steps of the processing of the patient’s history as well as to the writing and to the editing of the manuscript. TF, SS, US and SD have conceived the idea for the study and have written the first draft. SS has provided research support and advice throughout the project. Furthermore, CEG has provided expertise in artwork. All authors have read and approved the final version of the manuscript.
Delayed bone healing is an eminent problem in the operative treatment of distal tibia fractures. To address this problem from a biomechanical perspective, the DLS 3.7 (Dynamic Locking Screw 3.7 mm) as a new generation of locking screws has been developed. This screw enables the surgeon to control the rigidity of the plate osteosynthesis and thereby to expand clinical options in cases where the bridge plating is chosen for fracture treatment.
The purpose of the present prospective study was to evaluate the safety use of the DLS 3.7 in distal tibia fractures where bridge plating osteosynthesis is recommended.
In a prospective non-controlled cohort study, 34 patients with acute distal tibia fractures (AO 43 A-C) were treated with an angular stable plate fixation using DLS 3.7 or LHS 3.5. Follow-up examinations were performed three, six, twelve, and twenty-four weeks postoperatively and all registered complications were carefully collected.
A total of 34 patients were prospectively enrolled in this study with a minimum follow-up of 6 months or obvious osseous consolidation at an earlier stage. No complications directly related to the DLS 3.7 were recorded and no infections were observed.
This observational study could show that the DLS 3.7 in combination with locking compression plates provides a secure and easy application. According to the recent literature inter-fragmentary micro-motion is one evident goal to increase the reliability in fracture healing. The new DLS 3.7 with a maximum micro-motion of 0.2 mm combines the advantage of micro-motion with the well-known advantages of angular stable plate fixation.
Authors’ original file for figure 113037_2013_221_MOESM1_ESM.tif
Authors’ original file for figure 213037_2013_221_MOESM2_ESM.tiff
Authors’ original file for figure 313037_2013_221_MOESM3_ESM.tiff
Authors’ original file for figure 413037_2013_221_MOESM4_ESM.tiff
Authors’ original file for figure 513037_2013_221_MOESM5_ESM.jpeg
Bottlang M, Doornink J, Fitzpatrick DC, Madey SM: Far Cortical Locking Can Reduce Stiffness of Locked Plating Constructs While Retaining Construct Strength. J Bone Joint Surg Am. 2009, 91:
Goodship AE, Kenwright J: The influence of induced micromovement upon the healing of experimental tibial fractures. J Bone Joint Surg Br. 1985, 67: 650-655. PubMed
Klein P, Schell H, Streitparth F, Heller M, Kassi J-P, Kandziora F, Bragulla H, Haas NP, Duda GN: The initial phase of fracture healing is specifically sensitive to mechanical conditions. J Orthop Res. 2006, 21: 662-669. CrossRef
Wolf S, Janousek A, Pfeil J, Veith W, Haas F, Duda G, Claes L: The effects of external mechanical stimulation on the healing of diaphyseal osteotomies fixed by flexible external fixation. Clin Biomech (Bristol, Avon). 1998, 13: 359-364. 10.1016/S0268-0033(98)00097-7. CrossRef
Claes L, Heigele CA, Neidlinger-Wilke C, Kaspar D, Seidl W, Margevicius KJ, Augat P: Effects of mechanical factors on the fracture healing process. Clin Orthop Relat Res. 1998, 355: 132-S147. CrossRef
Stoffel K, Dieter U, Stachowiak G, Gächter A, Kuster MS: Biomechanical testing of the LCP – how can stability in locked internal fixators be controlled?. Injury. 2003, 34: 11-19. CrossRef
Doebele S, Horn C, Eichhorn S, Buchholtz A, Lenich A, Burgkart R, Nüssler AK, Lucke M, Andermatt D, Koch R, Stöckle U: The dynamic locking screw (DLS) can increase interfragmentary motion on the near cortex of locked plating constructs by reducing the axial stiffness. Langenbecks Arch Surg. 2010, 395: 421-428. 10.1007/s00423-010-0636-z. CrossRef
Schmal H, Strohm PC, Jaeger M, Südkamp NP: Flexible fixation and fracture healing: do locked plating “internal fixators” resemble external fixators?. J Orthop Trauma. 2011, 25 (Suppl 1): 15-20. CrossRef
Bottlang M, Feist F: Biomechanics of far cortical locking. J Orthop Trauma. 2011, 25 (Suppl 1): 21-28. CrossRef
Plecko M, Lagerpusch N, Andermatt D, Frigg R, Koch R, Sidler M, Kronen P, Klein K, Nuss K, Bürki A, Ferguson SJ, Stoeckle U, Auer JA, Rechenberg von B: The dynamisation of locking plate osteosynthesis by means of dynamic locking screws (DLS)-an experimental study in sheep. Injury. 2013, 44: 1346-1357. 10.1016/j.injury.2012.10.022. CrossRefPubMed
Kenwright J, Richardson JB, Cunningham JL, White SH, Goodship AE, Adams MA, Magnussen PA, Newman JH: Axial movement and tibial fractures. A controlled randomised trial of treatment. J Bone Joint Surg Br. 1991, 73 (4): 654-659. PubMed
Kershaw CJ, Cunningham JL, Kenwright J: Tibial external fixation, weight bearing, and fracture movement. Clin Orthop Relat Res. 1993, 293: 28-36. PubMed
Ahmad MA, Sivaraman A, Zia A, Rai A, Patel AD: Percutaneous locking plates for fractures of the distal tibia: our experience and a review of the literature. J Trauma Acute Care Surg. 2012, 72: E81-E87. PubMed
- Controlled dynamic stability as the next step in “biologic plate osteosynthesis” - a pilot prospective observational cohort study in 34 patients with distal tibia fractures
Christoph Emanuel Gonser
Yves P Acklin
- BioMed Central
Neu im Fachgebiet Chirurgie
Mail Icon II