Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Mesh fixation and broad overlap represent an open issue in umbilical hernia repair. A proprietary-designed implant with tentacle straps at its boundary has been developed to ensure a suture-free repair and a broader coverage of the abdominal wall. The study describes the results of umbilical hernia procedures carried out with the tentacle-shaped implant and the related surgical technique.
A proprietary tentacle-shaped flat mesh having a central body with integrated radiating arms at its edge was used to repair large umbilical hernias in 62 patients. The implant was placed in preperitoneal sublay. The friction of the straps, crossing the abdominal wall thanks to a special needle passer, was intended to assure adequate grip to hold the implant in place assuring a fixation-free procedure and broad overlap of the hernia defect.
In a mean follow-up of 48 months (range 10–62 months), 4 seromas and 2 ischemia of the navel skin occurred. No infections, hematomas, chronic pain, mesh dislocation, or recurrence has been reported.
The tentacle strap system of the prosthesis effectively ensured an easier implant placement avoiding the need for suturing the mesh. The arms of the implant ensured a proper orientation and stabilization of the mesh in association with a broad defect overlap. The specifically developed surgical procedure showed a quick postoperative recovery, a very low complication rate, and no recurrences even in the long term.
Musca AA (2014) Umbilical and ventral herniorrhaphy: a review of 1000 cases. Part 1: characteristics which predict surgical results and predisposing factors. Int Surg 48:169–179
Schumacher OP, Peiper C, Lorken M, Schumpelick V (2003) Langzeitergebnisse der Nabelhernienreparation nach Spitzy (Nahtverschluss). Chirurg 74:50–54
Bisgaard T, Kehlet H, Bay-Nielsen M, Iversen M, Rosenberg GJ, Jørgensen LN (2011) A nationwide study on readmission, morbidity, and mortality after umbilical and epigastric hernia repair. Hernia 15(5):541–546
Venclauskas L. Jokubauskas M, Zilinskas J, Zviniene K, Kiudelis M (2017) Long-term follow-up results of umbilical hernia repair. Wideochir Inne Tech Maloinwazyjne 12(4):350–356.
Amato G, Romano G, Goetze T, Salamone G, Agrusa A, Gulotta G, Paolucci V (2011) New mesh shape and improved implantation procedure to simplify and standardize open ventral hernia repair: a preliminary report. Hernia 15(6):659–665
Romano G, Calò PG, Erdas E, Medas F, Gordini L, Podda F, Amato G (2017) Fixation-free incisional hernia repair in the elderly: our experience with a tentacle-shaped implant. Aging Clin Exp Res 29(Suppl 1):173–177
Amato G, Romano G, Goetze T, Gordini L, Erdas E, Medas F, Calò P (2017) Long-Term Results of Fixation-Free Incisional Hernia Repair Using a Tentacle-Shaped Implant. Surg Technol Int 30:175–181
Altman D, Väyrynen T, Engh ME, Axelsen S, Falconer C (2008) For the Nordic Transvaginal Mesh Group. Short-term outcome after transvaginal mesh repair of pelvic organ prolapse. Int Urogynecol J Pelvic Floor Dysfunct 19(6):787–793
Abdel-Fattah M, Ramsay I (2008) West of Scotland Study Group. Retrospective multicentre study of the new minimally invasive mesh repair devices for pelvic organ prolapse. BJOG 115(1):22–30
Malik B, Lambaudie E, Collinet P, Dubois P, Cosson M (2007) Mechanical resistance of synthetic meshes for incontinence or prolapse surgery. Int Urogynecol J 18:183–187
Alcalay M, Livne M, Shidlovsky D, Hod E. (2009) Pullout force of polypropylene mesh deployed by endofast reliant fastener—a comparative study in a sheep model. ICS Congress Abstract 543
- Tentacle-shaped mesh for fixation-free repair of umbilical hernias
G. L. Canu
P. G. Calò
- Springer Paris
Neu im Fachgebiet Chirurgie
Mail Icon II